YY 07752010 (YY07752010) & related versions
Standard ID  Contents [version]  USD  STEP2  [PDF] delivered in  Standard Title (Description)  See Detail  Status  Similar PDF 
YY 07752010  English  150 
Add to Cart

09 seconds. Auto delivery.

Teleradiotherapy treatment planning system accuracy of dosimetric calculation and test methods for high energy X(γ) beam

YY 07752010
 
YY 07752010

Buy with any currencies (Euro, JPY, KRW...): YY 07752010 Preview this PDF: YY 07752010
YY 07752010: PDF in English YY 07752010 YY PHARMACEUTICAL INDUSTRY STANDARD OF THE PEOPLE’S REPUBLIC OF CHINA ICS 11.040.60 C 43 Teleradiotherapy treatment planning system accuracy of dosimetric calculation and test methods for high energy X (γ) beam 远距离放射治疗计划系统 ISSUED ON. DECEMBER 27, 2010 IMPLEMENTED ON. JUNE 1, 2012 Issued by. China Food and Drug Administration Table of Contents Foreword ... 3 1 Scope ... 4 2 Normative references ... 4 3 Terms and definitions ... 4 4 Requirements ... 5 4.1 Simple geometric conditions ... 5 4.2 Complex geometric conditions ... 5 4.3 Combination of complex geometric conditions ... 5 4.4 The outer edge of radiation field ... 6 4.5 The outer edge of radiation field, complex geometric conditions and the central axis is blocked ... 6 5 Test methods ... 6 5.1 General test conditions ... 6 5.2 Test example... 9 5.3 Simple geometric conditions ... 9 5.4 Complex geometric conditions ... 9 5.5 Combination of complex geometric conditions ... 10 5.6 The outer edge of radiation field ... 10 5.7 The outer edge of radiation field, complex geometric conditions and the central axis is blocked ... 10 Appendix A ... 11 Appendix B ... 12 References ... 23 Foreword All technical content of this Standard is mandatory. This Standard was drafted in accordance with the rules given in GB/T 1.12009. The test data and examples in this Standard adopt the test data (machine configuration data) and examples in IAEA No.1540 Technical Report. Please note that some of the content of this document may involve patents; the issuing agency of this document does not undertake the responsibility for the identification of these patents. This Standard was proposed by and shall be under the jurisdiction of Sub committee on Radiotherapy, Nuclear medicine and Radiation Dosimetry Equipment of National Technical Committee on Medical Electrical Equipment of Standardization Administration of China (SAC/TC 10/SC 3). Drafting organizations of this Standard. Beijing Institute for Medical Device Quality Supervision and Testing of State Food and Drug Administration. Main drafters of this Standard. Zhang Xin, Zhang Zhaoyuan, Yan Xu, Hu Jia and Qing Hou. Teleradiotherapy treatment planning system accuracy of dosimetric calculation and test methods for high energy X (γ) beam 1 Scope This Standard specifies the accuracy of dosimetric calculation and test methods of Radiotherapy Treatment Planning Systems (hereinafter referred to as RTPS). This Standard applies to RTPS with dosimetric calculation and display function of high energy X rays (4 MV ~ 25 MV), 60Co γ rays remote radiotherapy treatment on photon radiation. This Standard does not apply to the radiation treatment planning systems that are used by stereotactic radiotherapy, intensity modulated radiation therapy (IMRT) or other specialized radiotherapy techniques. However, the developers are encouraged to use this Standard in development, and use as reference standard in verifying dosimetric calculation algorithm. Test packets in this Standard cannot be used for clinical treatment planning. 2 Normative references The following referenced documents are indispensable for the application of this document. For dated references, only the edition cited applies. For undated references, the latest edition of the referenced document (including any amendments) applies. GB/T 178571999 Medical radiology  Terminology (Equipment for radiotherapy, nuclear medicine and radiation dosimetry) GB/T 189872003 Radiotherapy equipment  Coordinates, movements and scales 3 Terms and definitions The terms and definitions defined by GB/T 178571999 and the following ones apply to this document. The test requires to calculate point dose of a series of positions in a cube water body model. Users are recommended to specify a cube water body model of which the side length is 40 cm for RTPS, before test. Most RTSP allows to determine the dose measurement points by inputting coordinate values. However, if RTPS do not have the aforementioned functions, an easy way is to use beam’s central axis, offaxis distance (1cm, 2cm, 2.5cm, 3cm, 5cm, 6cm, 7cm, 9cm and 19cm) and calculation depth (dmax, 1cm, 3cm, 5cm, 10cm, 11cm, 15cm, 20cm, 25cm, 30cm and 35cm) to represent for each measurement point. Each test example has specified the aforementioned offaxis distance and depth. For each test example, the operator shall place radiation beam at the position stipulated in the test example; and mark the calculation point of depth and off axis distance required by test. It shall use a great enough radiation beam weighting to ensure that calculation result has enough accuracy, i.e., not using too small weighting may be because of that the rounding or data truncation causes the accuracy of calculation result less than 1%. 5.1.3 Beam dose setting In some RPTS, beam dose setting is corresponding to the dose when beam is transferred to dmax. While in some other systems, beam dose setting is corresponding to the dose when beam is transferred to reference radiation field size (i.e., 10 cm × 10 cm). Some other systems use different setting ways to set beam dose. Users shall be familiar with beam dose setting ways and ensure that calculation result is consistent to the conditions of obtaining corresponding test data. 5.1.4 Coordinate system In order to moreclearly express the geometric relationship between beam data and test examples, the following coordinate system is stipulated related to water body model. Water body model’s coordinate system is consistent to the fixed coordinate system stipulated in GB/T 189872003. Coordinate origin is at the isocenter of therapy equipment. Except isocenter test example (test example 3); isocenter of other test examples all locates at the surface of water body model. Zaxis is perpendicular to the upper surface and points upward from the surface. Zaxis coincides with the central axis of the radiation beam and points to radiation source. Oblique incidence test example is excluded. 5.5 Combination of complex geometric conditions Test shall conduct test example 12; calculate error according to formula (1). Test results shall comply with requirements of 4.3. 5.6 The outer edge of radiation field Test on measurement point of the outer edge of radiation field shall conduct test example 1, 2, 3, 6; calculate error according to formula (2). Where, δ2  Normalized relative error of measurement point of the outer edge of radiation field, %; Dc  Calculated values; Dr  Reference values; Da  Reference values of same depth on radiation beam. Test results shall comply with requirements of 4.4. 5.7 The outer edge of radiation field, complex geometric conditions and the central axis is blocked Test on central beam measurement point of the outer edge of radiation field under complex geometric conditions and the central axis is blocked shall conduct test example 5, 7; calculate error according to formula (3). Where, δ3  Normalized relative error of the outer edge of radiation field, complex geometric conditions and the central axis is blocked, %; Dc  Calculated values; Dr  Reference values; Do  Reference values of same depth in open field. Test results shall comply with requirements of 4.5. Appendix A (Normative) Test data This Standard introduces data package in IAEA No.1540 Technical Report [1] as predefined test conditions and test data of this Standard. The data package is divided into four parts, respectively including input data and test results of 6 MV, 10 MV, 18 MV and 60Co γ. When using data package, it shall select suitable data sets, input data by combining corresponding RTPS requirements, select appropriate radiation beam model, build up test’s geometric conditions, calculate dose of measurement point and input calculated values into evaluation table for error calculation. Test data package includes the following four parts.  IAEApack06MV NCS data package for 6 MV X;  IAEApack10MV NCS data package for 10 MV X;  IAEApack18MV NCS data package for ... ......
Standard ID  YY 07752010 (YY07752010)  Description (Translated English)  Teleradiotherapy treatment planning system accuracy of dosimetric calculation and test methods for high energy X(��) beam  Sector / Industry  Medical Device & Pharmaceutical Industry Standard  Classification of Chinese Standard  C43  Classification of International Standard  11.040.60  Word Count Estimation  22,263  Date of Issue  20101227  Date of Implementation  20120601  Quoted Standard  GB/T 178571999; GB/T 189872003  Drafting Organization  State Food and Drug Administration, Beijing Medical Device Quality Supervision and Inspection  Administrative Organization  National Standardization Technical Committee of Medical electrical equipment  Regulation (derived from)  State Food and Drug Administration Notice 2010 No. 97  Proposing organization  National Medical Electrical Standardization Technical Committee radiotherapy; nuclear Standardization Technical Committee on Medical science equipment and radiation dose (SAC/TC 10/SC 3)  Issuing agency(ies)  China Food and Drug Administration  Summary  This standard specifies the remote radiation treatment planning system (Radiotherapy Treatment Planning Systems, hereinafter referred to as RTPS) dose calculation accuracy requirements and test methods. This standard applies to medical highenergy Xray (4MV ~ 25MV), 60Co�� ray photon radiation has longrange radiation therapy dose calculation and display functions of RTPS, this standard does not apply to stereotactic radiotherapy and intensity modulated radiation therapy (IMRT) or Other specialized radiotherapy techniques used in radiation treatment planning system, but to encourage developers in the development and use of the application of the standard dose calculation algorithm as a validation reference standards. This standard test data packets can not be used in the clinical treatment planning. 
